High Temperature Glass Fiber Temperature Measuring Camera SystemDelivery term:The date of payment from buyers deliver within days
- Total supply:
- Delivery term:
The date of payment from buyers deliver within days
- Validity to:
- Last update:
- Browse the number:
- Changzhou Luming Photoelectric Technology Co., Ltd. By certification [File Integrity]
Research& Development Building, Information Industry Park, Science and Education City, Wujin District, Changzhou, Jiangsu, China
- Website: http://www.czlmcctv.com/ http://luming.sftechonline.com/
Product Name: High temperature glass fiber temperature measuring camera system
Product Model: LM-NK800
1. Furnace wall automatic type is suitable for glass fiber and other glass furnace.
2. Probe hood enter furnace wall, the thickness of furnace wall is no larger than 500millimetre.
3. Probe hood use stainless steel materials, with good corrosion resistance.
4. Special dustproof design, the lens does not gather dust.
5. Special water cooling, under the condition of poor water quality can normally use.
6. The high temperature pinhole zoom lens is ourselves develop and product, horizontal resolution is greater than 1.3 million pixels, under no cooling condition, it can withstand 350°C.
7. Probe hood can enter and exit automatically.
8. Using high-temperature air cylinder, with good reliability.
9. When water cooling or power supply have unforeseen circumstances, probe hood can exit automatically.
10. In the central control room or at site can control the equipment to enter and exit.
11. By DCS can control the equipment to enter and exit.
12. Using manual zoom high temperature pinhole lens can satisfy different furnace, different product requirements.
13. Using electric zoom high temperature pinhole lens can remote control the size and the clarity of image.
1. Probe will exit when the temperature of the hood rises above 55℃.
2. Probe will exit when the compressed air input pressure below 0.3MPa.
3. Probe will exit when the cooling water input flow is less than5L/min.
4. Probe will exit when the power is cut off.
5. Probe will exit when the control room or control device of site send out exit signal.
1. Lens definition: horizontal resolution is greater than 1.3 million pixels
2. Lens variable focusing: F4-F20 continuously adjustable
3. Exit distance: 500mm
1. Total intake pipe: G3/4”
2. Connecting pipe: 2-G1/2”(increase stop valve)
3. Inlet pressure: 0.4MPa～0.7MPa
4. Inlet temperature: ≤35°C
5. Minimum flow: 0.3M³/min
1. Total intake pipe: G1/2”
2. Connecting pipe: G1/2”
3. Inlet pressure: 0.1MPa～0.4MPa
4. Minimum flow: 30L/min
5. Inlet temperature: ≤35°C
Working Condition Temperature:
1. Control box: -5°C～75°C
2. Furnace temperature: ≦2000°C (continuous duty)
1. Automatically exit device: 1500*380*350 (unit: millimeter)
2. Control box: 700*1200*280 (width*height*thickness) (unit: millimeter)
3. Manipulator (there are three type, when ordering it, please specify)
3-1 Wall mounting (see figure)
3-2 Flush mounting (see figure)
3-3 Cabinet mounting or place on the operating desk (see figure)
4. Installation diagram: (see figure)
5. Open pore requirement (see figure)